How to Make a Digital Readout for Your Faceting Machine
A Digital Panel Meter (DPM) can measure angles to 0.01 degree if a precision instrument potentiometer is used and a stable reference voltage is used.
2 Minute Read
(Editor's Note: This is a first in a series of articles. Future articles will apply to specific machines, where this first one is general.)
A Digital Panel Meter (DPM) can measure angles to 0.01 degree if a precision instrument potentiometer is used and a stable reference voltage is used. This later version has much lower power consumption that the prior LED model, and can be run from batteries.
Principal of Operation
A trimpot and a sensing potentiometer form a simple voltage divider. Designing for 200 degrees of rotation, each degree equals one millivolt. In actual use, the critical measuring is done at that rotational quadrant from 0 to 90o, or from the Table to Girdle settings. Thus, if the pot is linear to 0.5% over its range, only one quarter of its range is really used. If batteries are used, rather than regulated power supplies, it only takes a minute to check the calibration once in a while. Lithium batteries seem to last forever in this application. The voltage divider only consumes 0.1 milliamp.
Less than 0.125% from zero to 90 degrees.
Cost to build:
Less than $50.
There is a critical
Retrofitting The Graves Faceter for a Digital Readout
Tin, Corian, and Ceramic Lap Testing
How to Score Polishing Laps
Ultra Tec Faceting Machine Factory
How Does Vapor Deposition Create Mystic Topaz and Aqua Aura Quartz?
Spurrite Value, Price, and Jewelry Information
Lapidary Technology Through the Ages: Laps and Polish
Why are Topaz and Citrine Gemstones Misidentified?
When you join the IGS community, you get trusted diamond & gemstone information when you need it.
Get started with the International Gem Society’s free guide to gemstone identification. Join our weekly newsletter & get a free copy of the Gem ID Checklist!